The SPICA-SAFARI instrument requires extremely sensitive transition edge sensor (TES) arrays with a noise equivalent power of 2×10-19W/Hz and a readout system with an output noise that is dominated by the detector noise. It is essential to ensure the frequency domain multiplexing (FDM) readout system in SAFARI meets the noise requirement. The FDM system in SAFARI consists essentially of LC filters, a superconducting quantum interference device, a room-temperature low-noise amplifier (LNA), and a demultiplexer. Here we present a noise study of the LNA from a laboratory amplifier chain. We found the equivalent current and voltage noise of the LNA to be 5.4pA/Hz and 315pV/Hz, respectively, which are low enough to read out SAFARI’s TES arrays.
|Number of pages||7|
|Journal||Journal of Low Temperature Physics|
|Publication status||Published - May-2020|
- Transition edge sensor
- Low-noise amplifier
- Readout noise